- Plate reader
Microplate Readers (also known as Plate readers) are laboratory instruments designed to detect biological, chemical or physical events of samples in microtiter plates. They are widely used in research, drug discovery, bioassay validation, quality control and manufacturing processes in the pharmaceutical and biotechnological industry and academic organizations. Sample reactions can be assayed in 6-1536 well format microtiter plates. The most common microplate format used in academic research laboratories or clinical diagnostic laboratories is 96-well (8 by 12 matrix) with a typical reaction volume between 100 and 200 µL per well. Higher density microplates (384- or 1536-well microplates) are typically used for screening applications, when throughput (number of samples per day processed) and assay cost per sample become critical parameters, with a typical assay volume between 5 and 50 µL per well. Common detection modes for microplate assays are absorbance, fluorescence intensity, luminescence, time-resolved fluorescence, and fluorescence polarization.
Absorbance detection has been available in microplate readers for more than 3 decades, and is used for assays such as ELISA assays, protein and nucleic acid quantification or enzyme activity assays (i.e. in the MTT assay for cell viability). A light source illuminates the sample using a specific wavelength (selected by an optical filter, or a monochromator), and a light detector located on the other side of the well measures how much of the initial (100 %) light is transmitted through the sample: the amount of transmitted light will typically be related to the concentration of the molecule of interest.
Fluorescence intensity detection has developed very broadly in the microplate format over the last two decades. The range of applications is much broader than when using absorbance detection, but instrumentation is usually more expensive. In this type of instrumentation, a first optical system (excitation system) illuminates the sample using a specific wavelength (selected by an optical filter, or a monochromator). As a result of the illumination, the sample emits light (it fluoresces) and a second optical system (emission system) collects the emitted light, separates it from the excitation light (using a filter or monochromator system), and measures the signal using a light detector such as a photomultiplier tube (PMT). The advantages of fluorescence detection over absorbance detection are sensitivity, as well as application range, given the wide selection of fluorescent labels available today. For example, a technique known as calcium imaging measures fluorescence intensity of calcium-sensitive dyes to asses intracellular calcium levels.
Luminescence detection is very popular for specific applications. The difference with fluorescence is that the light emitted by the samples is the result of a chemical or biochemical reaction (instead of being the result of excitation by light). Luminescence plate readers are simpler optically than fluorescence readers, as they don't require a light source, just a light detector. Typically, the optical system consists in a light-tight reading chamber, and PMT detector measuring the light emitted by the samples during the reaction. Common applications include luciferase-based gene expression assays, as well as cell viability and cytotoxicity assays based on the luminescent detection of ATP.
Time-resolved fluorescence (TRF)
Time-resolved fluorescence (TRF) measurement is very similar to fluorescence intensity (FI) measurement. The only difference is the timing of the excitation / measurement process. When measuring FI, the excitation and emission processes are simultaneous: the light emitted by the sample is measured while excitation is taking place. Even though emission systems are very efficient at removing excitation light before it reaches the detector, the amount of excitation light compared to emission light is such that FI measurements always exhibit fairly elevated background signals. TRF offers a solution to this issue. It relies on the use of very specific fluorescent molecules, called lanthanides, that have the unusual property of emitting over long periods of time (measured in milliseconds) after excitation, when most standard fluorescent dyes (e.g. fluorescein) emit within a few nanoseconds of being excited. As a result, it is possible to excite lanthanides using a pulsed light source (Xenon flash lamp or pulsed laser for example), and measure after the excitation pulse. This results in lower measurement backgrounds than in standard FI assays. The drawbacks are that the instrumentation and reagents are typically more expensive, and that the applications have to be compatible with the use of these very specific lanthanide dyes. The main use of TRF is found in drug screening applications, under a form called TR-FRET (time-resolved fluorescence energy transfer). TR-FRET assays are very robust (limited sensitivity to several types of assay interference) and are easily miniaturized. Robustness, the ability to automate and miniaturize are features that are highly attractive in a screening laboratory.
Fluorescence polarization measurement is also very close to FI detection. The difference is that the optical system includes polarizing filters on the light path: the samples in the microplate are excited using polarized light (instead of non-polarized light in FI and TRF modes). Depending on the mobility of the fluorescent molecules found in the wells, the light emitted will either be polarized or not. For example, large molecules (e.g. proteins) in solution, which rotate relatively slowly because of their size, will emit polarized light when excited with polarized light. On the other hand, the fast rotation of smaller molecules will result in a depolarization of the signal. The emission system of the plate reader uses polarizing filters to analyze the polarity of the emitted light. A low level of polarization indicates that small fluorescent molecules move freely in the sample. A high level of polarization indicates that fluorescent is attached to a larger molecular complex. As a result, one of the basic applications of FP detection is molecular binding assays, since they allow to detect if a small fluorescent molecule binds (or not) to a larger, non-fluorescent molecule: binding results in a slower rotation speed of the fluorescent molecule, and in an increase in the polarization of the signal.
There are instruments for measuring the dynamic or static light scattered from samples in a microplate. Companies that sell plate readers for dynamic light scattering include Wyatt Technology and Malvern Technology. Another company, Harbinger Biotechnology and Engineering, specializes in an instrument for static light scattering.
Many of the detection modes (absorbance, fluorescence intensity, luminescence, time-resolved fluorescence, and fluorescence polarization) are available stand-alone in dedicated plate readers, but are very often found today combined into one instrument (multi-mode plate reader). The range of applications for multi-mode plate readers is extremely large. Some of the most common assays are:
- Protein and cell growth assays
- Nucleic acid quantitation
- Molecular interactions
- Enzyme activity
- Cell toxicity, proliferation, and viability
- ATP quantification
- High throughput screening of compounds and targets in drug discovery
While "plate reader" usually refers to the devices described above, many variations are available. Some examples of other devices working with the microplate format are:
- ELISPOT plate readers, used to count the colored spots that are formed in the course of ELISPOT assays.
- High throughput imagers that can measure all the wells of a microplate at once
- High content screening (HCS) systems that image each well with high resolution, to look at cell populations
- Label-free instruments that use specialized microplates to measure binding events without the use of chemical markers
Laboratory equipment Glassware
Beaker · Boston round (bottle) · Büchner funnel · Burette · Cold finger · Condenser · Conical measure · Cuvette · Dean-Stark apparatus · Dropping funnel · Eudiometer · Evaporating dish · Gas syringe · Graduated cylinder · Pipette · Petri dish · Pycnometer · Separatory funnel · Soxhlet extractor · Watch glass
Agar plate · Aspirator · Autoclave · Biosafety cabinet · Bunsen burner · Calorimeter · Chemostat · Colony counter · Colorimeter · Laboratory centrifuge · Crucible · Eyewash · Fire blanket · Fume hood · Glove box · Homogenizer · Hot air oven · Incubator · Laminar flow cabinet · Magnetic stirrer · Meker-Fisher burner · Microscope · Microtiter plate · Picotiter plate · Plate reader · Retort stand · Safety shower · Spectrophotometer · Static mixer · Stir bar · Stirring rod · Scoopula · Teclu burner · Thermometer · Vacuum dry box · Vortex mixer · Wash bottleSee also Instruments used in medical laboratories
Wikimedia Foundation. 2010.