Distortionmeter (or more precisely distortion factor meter) is an electronic measuring instrument which displays the amount of distortion added to the original signal by an electronic circuit.
Distortion is equivalent to adding harmonics to a signal. When a purely sinusoidal signal is distorted, a series of harmonics is superimposed on the original signal.
If the input is
The normalized output is
The Total harmonics distortion (THD) is defined as the ratio of the harmonics to the fundamental;  i.e.,
This ratio can be given in dB or in percentage.
The distortionmeter is actually a levelmeter with two switchable parallel circuits at the input. The first circuit measures the total signal at the output of a system. (For low distortion levels this is almost equal to fundamental). That is calibrated to 100% or equivalently to 0 dB. The second circuit is a highpass filter which attenuates the fundamental, with negligible attenuation at harmonic frequencies. So when the instrument is switched for harmonics measurement the displayed level is the level of the harmonics only.
Analogue television broadcasting topics Systems Color systems VideoBack porch and front porch • Black level • Chrominance • Chrominance subcarrier • Colorburst • Color killer • Color TV • Composite video • Frame (video) • Horizontal scan rate • Horizontal blanking interval • Luma • Nominal analogue blanking • Overscan • Raster scan • Safe area • Television lines • Vertical blanking interval • White clipper Sound Modulation TransmissionAmplifiers • Antenna (radio) • Cavity amplifier • Differential gain • Differential phase • Diplexer • Dipole antenna • Dummy load • Frequency mixer • Intercarrier method • Intermediate frequency • Output power of an analog TV transmitter • Pre-emphasis • Residual carrier • Split sound system • Superheterodyne transmitter • Television receive-only • Television transmitter • Terrestrial television • Transmitter station • Transposer Frequencies & Bands Propagation TestingDistortionmeter • Field strength meter • Vectorscope • VIT signals • Zero reference pulse Artifacts Electrical and electronic measuring equipment
Ammeter · Capacitance meter · Distortionmeter · Electric energy meter · LCR meter · Microwave power meter · Multimeter · Network analyzer · Ohmmeter · Oscilloscope · Psophometer · Q meter · Signal analyzer · Signal generator · Spectrum analyzer · Transistor tester · Tube tester · Wattmeter · Vectorscope · Video signal generator · Voltmeter · VU meter
Wikimedia Foundation. 2010.