- Point spread function
The

**point spread function**(**PSF**) describes the response of an imaging system to apoint source or point object. A related but more general term for the PSF is a system'simpulse response . The PSF in many contexts can be thought of as the extended blob in an image that represents an unresolved object. In functional terms it is thespatial domain version of themodulation transfer function . It is a useful concept inFourier optics , astronomical imaging, electron microscopy and other imaging techniques such as 3Dmicroscopy (like inConfocal laser scanning microscopy ) andfluorescence microscopy . The degree of spreading (blurring) of the point object is a measure for the quality of an imaging system. In incoherent imaging systems such asfluorescent microscopes ,telescopes or optical microscopes, the image formation process is linear in power and described bylinear system theory. When the light is coherent, image formation is linear in complex field.vague This means that when two objects A and B are imaged simultaneously, the result is equal to the sum of the independently imaged objects. In other words: the imaging of A is unaffected by the imaging of B and "vice versa", owing to the non-interacting property of photons. (The sum is of the light waves which may result in destructive and constructive interference at non-image planes.)**Introduction**By virtue of the linearity property of optical imaging systems, i.e.,

: "Image"("Object"

_{1}+ "Object"_{2}) = "Image"("Object"_{1}) + "Image"("Object"_{2})the image of an object in a microscope or telescope can be computed by expressing the object-plane field as a weighted sum over 2D impulse functions, and then expressing the image plane field as the weighted sum over the "images" of these impulse functions. This is known as the "superposition principle", valid for

linear systems . The images of the individual object-plane impulse functions are called point spread functions, reflecting the fact that a mathematical "point" of light in the object plane is "spread" out to form a finite area in the image plane (in some branches of mathematics and physics, these might be referred to asGreen's functions orimpulse response functions).When the object is divided into discrete point objects of varying intensity, the image is computed as a sum of the PSF of each point. As the PSF is typically determined entirely by the imaging system (that is, microscope or telescope), the entire image can be described by knowing the optical properties of the system. This process is usually formulated by a

convolution equation. Inmicroscope image processing andastronomy , knowing the PSF of the measuring device is very important for restoring the (original) image withdeconvolution .**Theory**The point spread function may be independent of position in the object plane, in which case it is called "shift invariant". In addition, if there is no distortion in the system, the image plane coordinates are linearly related to the object plane coordinates via the

magnification "M" as::$(x\_i,\; y\_i)\; =\; (M\; x\_o,\; M\; y\_o)$.

If the imaging system produces an inverted image, we may simply regard the image plane coordinate axes as being reversed from the object plane axes. With these two assumptions, i.e., that the PSF is shift-invariant "and" that there is no distortion, calculating the image plane convolution integral is a straightforward process.

Mathematically, we may represent the object plane field as:

:$O(x\_o,y\_o)\; =\; int!!int\; O(u,v)\; ~\; delta(x\_o-u,y\_o-v)\; ~\; du,\; dv$

i.e., as a sum over weighted impulse functions, although this is also really just stating the sifting property of 2D delta functions (discussed further below). Rewriting the object transmittance function in the form above allows us to calculate the image plane field as the superposition of the images of each of the individual impulse functions, i.e., as a superposition over weighted point spread functions in the image plane using the "same" weighting function as in the object plane, i.e., $O(x\_o,y\_o)$. Mathematically, the image is expressed as:

:$I(x\_i,y\_i)\; =\; int!!int\; O(u,v)\; ~\; mathrm\{PSF\}(x\_i\; -\; M\; u,y\_i\; -\; M\; v)\; ,\; du,\; dv$

in which PSF("x

_{i}" − "Mu","y_{i}" − "Mv") is the image of the impulse function δ("x_{o}" − "u","y_{o}" − "v").The 2D impulse function may be regarded as the limit (as side dimension "w" tends to zero) of the "square post" function, shown in the figure below (click to enlarge).

We imagine the object plane as being decomposed into square areas such as this, with each having its own associated square post function. If the height, "h", of the post is maintained at 1/w

^{2}, then as the side dimension "w" tends to zero, the height, "h", tends to infinity in such a way that the volume (integral) remains constant at 1. This gives the 2D impulse the sifting property (which is implied in the equation above), which says that when the 2D impulse function, δ("x" − "u","y" − "v"), is integrated against any other continuous function, "f"("u","v"), it "sifts out" the value of "f" at the location of the impulse, "i"."e"., at the point ("x","y").Since the concept of a perfect point source object is so central to the idea of PSF, it's worth spending some time on that before proceeding further. First of all, there is no such thing in nature as a perfect mathematical point source radiator; the concept is completely non-physical and is nothing more that a mathematical construct used to model and understand optical imaging systems. The utility of the point source concept comes from the fact that a point source in the 2D object plane can only radiate a perfect uniform-amplitude, spherical wave - a wave having perfectly spherical, outward travelling phase fronts with uniform intensity everywhere on the spheres (see

Huygens-Fresnel principle ). Such a source of uniform spherical waves is shown in the figure below (click to enlarge). We also note that a perfect point source radiator will not only radiate a uniform spectrum of propagating plane waves, but a uniform spectrum of exponentially decaying (evanescent) waves as well, and it is these which are responsible for resolution finer than one wavelength (seeFourier optics ). This follows from the followingFourier transform expression for a 2D impulse function,:$delta\; (x,y)\; propto\; int!!int\; e^\{j(k\_x\; x\; +\; k\_y\; y)\}\; ,\; d\; k\_x,\; d\; k\_y$

The quadratic lens intercepts a "portion" of this spherical wave (and the key phrase here is "a portion"), and refocuses it onto a blurred point in the image plane. For a single lens, an on-axis point source in the object plane produces an

Airy disc PSF in the image plane. This comes about in the following way. "It can be shown" (seeFourier optics ,Huygens-Fresnel principle ,Fraunhofer diffraction ) that the field radiated by a planar object (or, by reciprocity, the field converging onto a planar image) is related to its corresponding source (or image) plane distribution via aFourier transform (FT) relation. In addition, a uniform function over a circular area (in one FT domain) corresponds to the Airy function, "J"_{1}("x")/"x" in the other FT domain, where "J"_{1}("x") is the first-order Bessel function of the first kind. That is, a uniformly-illuminated circular aperture radiates an Airy function pattern plus a converging uniform spherical wave (over a circular angular sector) and yields an Airy function image. A graph of a sample 2D Airy function is shown in the adjoining figure (click to enlarge).Therefore, the converging ("partial") spherical wave shown in the figure above produces an

Airy disc in the image plane. The argument of the Airy function is important, because this determines the "scaling" of the Airy disc (in other words, how big the disc is in the image plane). If Θ_{max}is the maximum angle that the converging waves make with the lens axis, "r" is radial distance in the image plane, andwavenumber "k" = 2π/λ where λ = wavelength, then the argument of the Airy function is: kr sin(Θ_{max}). If Θ_{max}is small (only a small portion of the converging spherical wave is available to form the image), then radial distance, r, has to be very large before the total argument of the Airy function moves away from the central spot. In other words, if Θ_{max}is small, the Airy disc is large (which is just another statement of Heisenberg'suncertainty principle for FT pairs, namely that small extent in one domain corresponds to wide extent in the other domain, and the two are related via the "space-bandwidth product". By virtue of this, highmagnification systems, which typically have small values of Θ_{max}(by theAbbe sine condition ), can have more blur in the image, owing to the broader PSF. The size of the PSF is proportional to themagnification , so that the blur is no worse in a relative sense, but it is definitely worse in an absolute sense.**History and methods**The diffraction theory of point-spread functions was first studied by Airy in the nineteenth century. He developed an expression for the point-spread function amplitude and intensity of a perfect instrument, free of aberrations (the so-called

Airy disc ). The theory of aberrated point-spread functions close to the optimum focal plane was studied by the Dutch physicistsFritz Zernike and Nijboer in the 1930–40s. A central role in their analysis is played by Zernike’s circle polynomials that allow an efficient representation of the aberrations of any optical system with rotational symmetry. Recent analytic results have made it possible to extend Nijboer and Zernike’s approach for point-spread function evaluation to a large volume around the optimum focal point. This [*http://www.nijboerzernike.nl Extended Nijboer-Zernike (ENZ) theory*] is instrumental in studying the imperfect imaging of three-dimensional objects inconfocal microscopy or astronomy under non-ideal imaging conditions. The ENZ-theory has also been applied to the characterization of optical instruments with respect to their aberration by measuring the through-focus intensity distribution and solving an appropriateinverse problem .**PSF in microscopy**In microscopy, experimental determination of a PSF is usually tricky, due to the difficulty of finding sub-resolution (point-like) radiating sources.

Quantum dot s andfluorescent bead s are usually considered for this purpose.**The PSF in astronomy**In

observational astronomy the experimental determination of a PSF is often very straightforward due to the ample supply of point sources (star s orquasars ). The form and source of the PSF may vary widely depending on the instrument and the context in which it is used.For

radio telescopes and diffraction-limited spacetelescopes the dominant terms in the PSF may be inferred from the configuration of the aperture in theFourier domain . In practice there may be multiple terms contributed by the various components in a complex optical system. A complete description of the PSF will also include diffusion of light (or photo-electrons) in the detector, as well as tracking errors in the spacecraft.For ground based optical telescopes, atmospheric turbulence (known as

astronomical seeing ) dominates the contribution to the PSF. In high-resolution ground-based imaging, the PSF is often found to vary with position in the image (an effect called anisoplanatism). In ground basedadaptive optics systems the PSF is a combination of the aperture of the system with residual uncorrected atmospheric terms.**Point spread functions in ophthalmology**PSFs have recently become a useful diagnostic tool in clinical

ophthalmology . Patients are measured with awavefront sensor, and special software calculates the PSF for that patient’s eye. In this manner a physician can "see" what the patient sees. This method also allows a physician to simulate potential treatments on a patient, and see how those treatments would alter the patient’s PSF. Additionally, once measured the PSF can be minimized using an adaptive optics system. This, in conjunction with a CCD, can be used to visualize anatomical structures not otherwise visible "in vivo", such as cone photoreceptors.**ee also***

Circle of confusion , for the closely related topic in general photography.

*Airy disc **External links*** [

*http://support.svi.nl/wiki/NyquistCalculator PSF calculator*] for fluorescence microscopes.

* [*http://www.nijboerzernike.nl Extended Nijboer-Zernike (ENZ) theory*] , calculating the PSF for a general Optical system.

*Wikimedia Foundation.
2010.*