# Mass flow meter

Mass flow meter
A mass flow meter

A mass flow meter, also known as an inertial flow meter is a device that measures mass flow rate of a fluid traveling through a tube. The mass flow rate is the mass of the fluid traveling past a fixed point per unit time.

The mass flow meter does not measure the volume per unit time (e.g., cubic meters per second) passing through the device; it measures the mass per unit time (e.g., kilograms per second) flowing through the device. Volumetric flow rate is the mass flow rate divided by the fluid density. If the density is constant, then the relationship is simple. If the fluid has varying density, then the relationship is not simple. The density of the fluid may change with temperature, pressure, or composition, for example. The fluid may also be a combination of phases such as a fluid with entrained bubbles.

## Operating principle of a coriolis flow meter

There are two basic configurations of coriolis flow meter: the curved tube flow meter and the straight tube flow meter. This article discusses the curved tube design.

 A rotating mass flow meter to illustrate the operating principle of the vibrating mass flow meter. double sized version When fluid is flowing through the tubes, the tubes twist slightly. double sized version

The animations on the right do not represent an actually existing coriolis flow meter design. The purpose of the animations is to illustrate the operating principle, and to show the connection with rotation.

Fluid is being pumped through the mass flow meter. When there is mass flow, the tube twists slightly. The arm through which fluid flows away from the axis of rotation must exert a force on the fluid, to increase its angular momentum, so it bends backwards. The arm through which fluid is pushed back to the axis of rotation must exert a force on the fluid to decrease the fluid's angular momentum again, hence that arm will bend forward.

In other words, the inlet arm is lagging behind the overall rotation, and the outlet arm leads the overall rotation.

The principle design of a curved tube mass flow meter. The vibration pattern during no-flow.
double sized version

The animation on the right represents how curved tube mass flow meters are designed. When the fluid is flowing, it is led through two parallel tubes. An actuator (not shown) induces a vibration of the tubes. The two parallel tubes are counter-vibrating, to make the measuring device less sensitive to outside vibrations. The actual frequency of the vibration depends on the size of the mass flow meter, and ranges from 80 to 1000 vibrations per second. The amplitude of the vibration is too small to be seen, but it can be felt by touch.

When no fluid is flowing, the vibration of the two tubes is symmetrical, as shown in the animations.

Vibration pattern of the tubes when there is mass flow.
double sized version

The animation on the right represents what happens during mass flow. When there is mass flow, there is some twisting of the tubes. The arm through which fluid flows away from the axis of rotation must exert a force on the fluid to increase its angular momentum, so it is lagging behind the overall vibration. The arm through which fluid is pushed back towards the axis of rotation must exert a force on the fluid to decrease the fluid's angular momentum again, hence that arm leads the overall vibration.

The inlet arm and the outlet arm vibrate with the same frequency as the overall vibration, but when there is mass flow the two vibrations are out of sync, the inlet arm is behind, the outlet arm is ahead. The two vibrations are shifted in phase with respect to each other, and the degree of phase-shift is a measure for the amount of mass that is flowing through the tubes.

## Density and volume measurements

The mass flow of a u-shaped coriolis flow meter is given as: $Q_m=\frac{ K_u -I_u\omega^2 }{2Kd^2}\tau$

where Ku is the temperature dependent stiffness of the tube, K a shape-dependent factor, d the width, τ the time lag, ω the vibration frequency and Iu the inertia of the tube. As the inertia of the tube depend on its contents, knowledge of the fluid density is needed for the calculation of an accurate mass flow rate.

If the density changes too often for manual calibration to be sufficient, the coriolis flow meter can be adapted to measure the density as well. The natural vibration frequency of the flow tubes depend on the combined mass of the tube and the fluid contained in it. By setting the tube in motion and measuring the natural frequency, the mass of the fluid contained in the tube can be deduced. Dividing the mass on the known volume of the tube gives us the density of the fluid.

Such an instantaneous density measurement in turn allow us to calculate the flow in volume per time, by dividing mass flow with density.

## Calibration

Both mass flow and density measurements depend on the vibration of the tube. This depends on the rigidity of the tube which in turn depend on its temperature. Calculations must therefore take the temperature of the fluid into account.

Wikimedia Foundation. 2010.

### Look at other dictionaries:

• Mass flow rate — is the mass of substance which passes through a given surface per unit time. Its unit is mass divided by time, so kilogram per second in SI units, and slug per second or pound per second in US customary units. It is usually represented by the… …   Wikipedia

• Mass flow controller — A mass flow controller A mass flow controller (MFC) is a device used to measure and control the flow of fluids and gases. A mass flow controller is designed and calibrated to control a specific type of fluid or gas at a particular range of flow… …   Wikipedia

• mass airflow meter — (MAF) device for measuring the mass flow of air into an engine. Also see airflow sensor hot wire airflow meter …   Dictionary of automotive terms

• Mass flow sensor — A mass airflow sensor. A mass air flow sensor is used to find out the mass flowrate of air entering a fuel injected internal combustion engine. The air mass information is necessary for the engine control unit (ECU) to balance and deliver the… …   Wikipedia

• Air flow meter — Contents 1 Types of air flow meters 1.1 In automobiles 1.1.1 Failures …   Wikipedia

• air flow meter — A meter which measures the rate at which air enters the engine. In Bosch systems, any device that measures the amount of air being used by the engine. The control unit uses this information to determine the load on the engine. The two most common …   Dictionary of automotive terms

• Flow measurement — is the quantification of bulk fluid movement. Flow can be measured in a variety of ways. Positive displacement flow meters acumulate a fixed volume of fluid and then count the number of times the volume is filled to measure flow. Other flow… …   Wikipedia

• Flow sensor — A flow sensor is a device for sensing the rate of fluid flow. Typically a flow sensor is the sensing element used in a flow meter, or flow logger, to record the flow of fluids.As is true for all sensors, absolute accuracy of a measurement… …   Wikipedia

• flow — flowable, adj. flowability, n. /floh/, v.i. 1. to move along in a stream: The river flowed slowly to the sea. 2. to circulate: blood flowing through one s veins. 3. to stream or well forth: Warmth flows from the sun. 4. to issue or proceed from a …   Universalium

• meter — [1] A device for measuring the quantity of a substance passing through it. [2] To measure with a meter. [3] to supply at a measured rate. [4] A distance of about a yard (actually a little over 39.37 ). Also spelled metre. Also see airflow meter… …   Dictionary of automotive terms