# Magnetic level gauge

Magnetic level gauge

A magnetic level gauge is used to measure the level of fluids. A magnetic level gauge includes a “floatable” device that can float both in high and low density fluids. Magnetic level gauges may also be designed to accommodate severe environmental conditions up to 210 bars at 370°C.

## Magnetic level sensors

Magnetic float level sensors involve the use of a permanent magnet sealed inside a float whose rise and fall causes the opening or closing of a mechanical switch, either through direct contact or in proximity of a reed switch. With mechanically-actuated floats, the float is directly connected to a micro switch. For both magnetic and mechanical float level sensors, chemical compatibility, temperature, specific gravity (density), buoyancy, and viscosity affect the selection of the stem and the float. For example, larger floats may be used with liquids with specific gravities as low as 0.5 while still maintaining buoyancy. The choice of float material is also influenced by temperature-induced changes in specific gravity and viscosity - changes that directly affect buoyancy.

## Explanation

In order to explore the physics and engineering behind this design, we need to take a look at basic magnetism. If we look at a standard bar magnet, there are two magnetic poles - north and south. (The north will read positive on a gauss meter and the south will read negative.) In order to map a magnetic field, we look at magnetic flux lines. Magnetic flux lines are a graphical representation of the magnetic field density. They show the direction of flow for the magnetic field and represent relative field strength - the closer together the lines are, the stronger the magnetic field. Flux lines will always travel from the north pole to the nearest south pole and always leave and enter surfaces at 90°, or perpendicular to the surface. They can only travel in straight lines or curved paths, which means they can never make a sudden, abrupt change in direction. Flux lines will also always follow a path of least magnetic resistance. Most importantly, they can never cross one another.

## Considerations

When selecting a magnetic level gauge it is important to take into account the strength of the magnetic field. The magnetic field is the heart of the magnetic level gauge – the stronger the field, the more reliable the instrument will function. Some manufacturers rely on a single magnet for their magnetic level gauges which causes the strength of the north field to be identical to, and as weak as, the south field. It is apparent that at the location of the indicators, switches and transmitters, the field would not be as intense. Some manufacturers use a single annular ring magnet, others use a series of single bar magnets in a circular array in their float design. In this design the relative field strength of the north and south poles will be equal to one another and less than that of a dual magnet design. Moreover, the field strength as you travel around the circumference will have high and low spots as you pass between the individual bar magnets.

Wikimedia Foundation. 2010.

### Look at other dictionaries:

• Magnetic base — as used in the engineering metaltrades. Fixed magnetic base with sna …   Wikipedia

• Level sensor — Level sensors are used to detect liquid level. The liquid to be measured can be inside a container or can be in its natural form (e.g. a river or a lake). The level measurement can be either continuous or point values. Continuous level sensors… …   Wikipedia

• Magnetic catalysis — is a phenomenon in quantum field theory which explains a spontaneous breaking of flavor or chiral symmetry, triggered by the presence of an external magnetic field. Contents 1 General description 2 Possible applications 2.1 Chiral symmetry brea …   Wikipedia

• Introduction to gauge theory — This article is an accessible, non technical introduction to the subject. For the main encyclopedia article, see Gauge theory. Quantum field theory …   Wikipedia

• Marking gauge — Stanley and Veritas marking gauges A marking gauge, also known as a scratch gauge,[1] is used in woodworking and metalworking to mark out lines for cutting or other operations.[2] …   Wikipedia

• List of sensors — * Accelerometer * Touch sensor * Active pixel sensor * Air flow meter * Alarm sensor * Bedwetting alarm * Bhangmeter * Biochip * Biosensor * Breathalyzer * Capacitance probe * Carbon paste electrode * Carbon monoxide detector * Catadioptric… …   Wikipedia

• hydrologic sciences — Introduction       the fields of study concerned with the waters of the Earth. Included are the sciences of hydrology, oceanography, limnology, and glaciology.       In its widest sense hydrology encompasses the study of the occurrence, the… …   Universalium

• Inclinometer — Compass with inclinometer Military model An inclino …   Wikipedia

• Liquid capacitive inclinometers — are Inclinometers (or Clinometers)whose sensing elements are made with liquid capacitive technology; they are considered gravity based. The best way to understand how they work is to think of a disc like cavity that is half filled with a… …   Wikipedia

• Komelon — Co. Ltd. (hangul:코멜론주식회사, kosdaq|049430) is a Korea s chemical, machinery, tool and engineering company. headquartered in Gosan dong Namdong gu Incheon, Korea. established in 1963. it was a manufactured in plastic and steel tape, level gauge and… …   Wikipedia