Actigraphy is a relatively
non-invasivemethod of monitoring human rest/activity cycles. A small actigraph unit, also called an actimetry sensor, [cite journal | last = Pigot | first = Hélène | coauthors = Bernard Lefebvre, Jean-Guy Meunier, Brigitte Kerhervé, André Mayers, Sylvain Giroux | year = 2003 | title = The role of intelligent habitats in upholding elders in residence | journal = | pages = | publisher = Département de mathématiques et d'informatique, Université de Sherbrooke | location = Canada | url = http://www.dmi.usherb.ca/~sgiroux/Publications/2003/2003-Biomedecine-Pigot-01.pdf | format = PDF | accessdate = 2008-01-22 | quote = ] is worn by a patient to measure gross motor activity. Motor activity often under test is that of the wrist, measured by an actigraph in a wrist-watch-like package. The unit continually records the movements it undergoes. The data is later read to a computer where it can be analysed.
Sleep actigraphs are generally watch-shaped and worn on the wrist of the non-dominant arm. They are useful for determining sleep patterns and
circadian rhythms. They may be worn for weeks at a time.
Actigraphy is useful for assessing daytime sleepiness in situations where a laboratory sleep latency test is not appropriate. It is used to clinically evaluate
insomnia, circadian rhythm sleep disorders, excessive sleepinessand restless legs syndrome. It is also used in assessing the effectiveness of pharmacologic, behavioural, phototherapeutic or chronotherapeutic treatments for such disorders.
Actigraphy has not traditionally been used in routine diagnosis of sleep disorders but is increasingly being employed in sleep clinics to replace full
polysomnography. The technique is more extensively used in academic research and is being increasingly employed in new drug clinical trials where sleep quality is seen as a good indicator of quality of life.
Activity actigraphs are worn and used similar to a
pedometer: around the waist, near the hip. They are useful for determining the amount of activity and possibly the number of calories burned by the wearer. They are worn for a number of days.
Movement actigraphs are generally larger and worn on the shoulder of the dominant arm. They contain a 3D actigraph as opposed to a single dimension one, and have a high sample rate and a large memory. They are used for only a few hours, and can be used to determine problems with gait and other physical impairments.
The actigraph unit
The unit itself is an electronic device which generally consists of
* a piezoelectric
low-pass filterwhich filters out everything except the 2-3 Hz band, thereby ensuring external vibrations are ignored,
* a timer to start/stop the actigraph at specific times, and to accumulate values for a specific time frame,
* a memory to store the resulting values, and
* an interface, usually
USBor serial, to program the timer and download the data from memory.
Actigraphs have a number of different ways of accumulating the values from the accelerometer in memory. ZCM (zero crossing mode) counts the number of times the accelerometer waveform crosses 0 for each time period. PIM (proportional integral mode) measures the area under the curve, and adds that size for each time period. TAT (time above threshold) uses a certain threshold, and measures the length of time that the wave is above a certain threshold. Literature shows that PIM provides most accurate measurements for both sleep and activity, though the difference with ZCM is marginal.
Actigraph units vary widely in size and features and can be expanded to include additional measurements. However, there are a number of limiting factors:
* Fastest sample rate: 1 minute intervals provide adequate detail to measure sleep, but could be too slow for measuring other parameters.
* Amount of memory: Together with sample rate, the amount of memory determines how long measurements can be taken.
* Battery usage: Some actigraphs have a short battery life.
* Weight: the heavier the actigraph, the more invasive its use.
* Water resistance: for proper measurements it is often desirable that the actigraph be worn in the shower, bathtub, or even while swimming/diving.
For some uses, the following are examples of additional features:
* Watch functionality: making the device more attractive to the user.
* User input: most actigraphs now include a button so the user can indicate a specific event that occurs, for example lights out at bedtime.
* Subjective user input: for example a query function to allow surveys at specific times.
* Sensors which monitor:
** ambient light
** sound levels
** parkinsonian tremor
** skin resistance
** a full EEG data stream
List of actigraphs
* [http://actiwatch.respironics.com/] , Philips Respironics Actiwatch
* [http://www.camntech.com/h_aw_1.html Cambridge Neurotechnology Actiwatch]
* [http://www.ambulatory-monitoring.com/ Ambulatory Monitoring Inc.] , Motionlogger Actigraph
* [http://www.theactigraph.com/ ActiGraph LLC, ActiTrainer] (was CSA/MTI)
* [http://www.gcdataconcepts.com/products.html Gulf Coast Data Concepts] USB Accelerometer
* [http://www.cebrasil.com.br/Produto/ACT10.html Consultoria Eletronica] Four Parameter Brazilian Actigraph
* [http://www.aasmnet.org/Resources/PracticeParameters/PP_Actigraphy_Circ.pdf American Academy of Sleep Medicine] - Practice parameters for the role of actigraphy in the study of sleep and circadian rhythms: An update for 2002
* [http://actiwatch.respironics.com/ Philips Respironics] - Manufacturer of activity monitors
* [http://www.ambulatory-monitoring.com Ambulatory Monitoring, Inc.] - Maker of activity monitors
* [http://www.paltechnologies.com PAL Technologies Ltd] - Manufacturer of activity monitors
* [http://www.camntech.co.uk Cambridge Neurotechnology] - Manufacturer of activity monitors
* [http://www.acsm-msse.org/pt/re/msse/abstract.00005768-200409000-00026.htm;jsessionid=EGV2oXf1VwE3h24pSzos7mnaSrNuqJu7tco1AuGHNjH4WGODCzwO!987057721!-949856145!9001!-1?index=1&results=1&count=10&searchid=1&nav=search%20 Reliability of Accelerometry-Based Activity Monitors: A Generalizability Study. Gregory J. Welk, Jodee A. Schaben, and James R. Morrow, Jr. Medicine & Science In Sports & Exercise, Vol. 36, No. 9, PP. 1637-1645, 2004.] - Medicine & Science in Sports & Medicine, the official Journal of the American College of Sports Medicine
Wikimedia Foundation. 2010.