- Geiger-Müller tube
A Geiger-Müller tube (or GM tube) is the sensing element of a
Geiger counterinstrument that can detect a single particle of ionizing radiation, and typically produce an audible click for each. It was named for Hans Geigerwho invented the device in 1908, and Walther Müllerwho collaborated with Geiger in developing it further in 1928. [cite journal
title = Elektronenzählrohr zur Messung schwächster Aktivitäten
doi = 10.1007/BF01494093
author = H. Geiger and W. Müller
volume = 16
issue = 31
year = 1928
journal = Naturwissenschaften | pages = 617] It is a type of
gaseous ionization detectorwith an operating voltage in the Geiger plateau.
The Geiger counter is sometimes used as a
hardware random number generator.
Description and operation
A Geiger-Müller tube consists of a tube filled with an low-pressure (~0.1 Atm) inert gas such as
helium, neonor argon, in some cases in a Penning mixture, and an organic vapor or a halogen gas and contains electrodes, between which there is a voltage of several hundred volts, but no current flowing. The walls of the tube are either metal or the inside coated with metal or graphiteto form the cathodewhile the anodeis a wirepassing up the center of the tube.
ionizing radiationpasses through the tube, some of the gas molecules are ionized, creating positively charged ions, and electrons. The strong electric field created by the tube's electrodes accelerates the ions towards the cathode and the electrons towards the anode. The ion pairs gain sufficient energy to ionize further gas molecules through collisions on the way, creating an avalanche of charged particles.
This results in a short, intense pulse of current which passes (or "cascades") from the negative electrode to the positive electrode and is measured or counted.
Most detectors include an
audio amplifierthat produce an audible click on discharge. The number of pulses per secondmeasures the intensity of the radiation field. Some Geiger counters display an exposure rate (e.g. mR·h), but this does not relate easily to a dose rate as the instrument does not discriminate between radiation at different energy
The usual form of tube is an "end-window tube". This type is so-named because the tube has a window at one end through which ionizing radiation can easily penetrate. The other end normally has the electrical connectors. There are two types of end-window tubes: the
glass-mantle type and the micawindow type. The glass window type will not detect alpha radiation since it is unable to penetrate the glass, but is usually cheaper and will usually detect beta radiationand X-rays. The mica window type will detect alpha radiation but is more fragile.
Most tubes will detect
gamma radiation, and usually beta radiationabove about 2.5 MeV. Geiger-Müller tubes will not normally detect neutrons since these do not ionise the gas. However, neutron-sensitive tubes can be produced which either have the inside of the tube coated with boronor contain boron trifluorideor helium-3gas. The neutrons interact with the boron nuclei, producing alpha particles or with the helium-3 nuclei producing hydrogen and tritiumions and electrons. These charged particles then trigger the normal avalanche process.
The G.M. tube must produce a single pulse on entry of a single particle.It must not give any spurious pulse and recover quickly to the passive state.But unfortunately the positive Ar ions that eventually strike the cathode become neutral Ar atoms in an excited state by gaining electrons from the cathode. The excited atoms return to the ground state by emitting photons and these photons cause avalanches and hence spurious pulses.
To prevent the current from flowing continuously there are several techniques to stop, or quench the discharge. Quenching is important because a single particle entering the tube is counted by a single discharge, and so it will be unable to detect another particle until the discharge has been stopped, and because the tube is damaged by prolonged discharges.
"External quenching" uses external electronics to remove the high voltage between the electrodes. "Self-quenching" or "internal-quenching" tubes stop the discharge without external assistance, and contain a small amount of a polyatomic organic vapor such as
butaneor ethanol; or alternatively a halogensuch as bromineor chlorine.
If the diatomic gas(quencher) is introduced in the tube, the positive Ar ions, during their slow motion to the cathode, would have multiple collisions with the quencher gas molecules and transfer their charge and some energy to them. Thus neutral Ar atoms would reach the cathode. The quencher gas ions in their turn reach the cathode, gain electrons thereform and move into excited states. But these excited molecules lose their energy not by photon emission but by dissociation into neutral quencher molecules. [Atomic and Nuclear Physics by Gupta & Ghosh] No spurious pulses are thus produced.
Invention of halogen tubes
The halogen tubes were invented by
Sidney H. Liebsonin 1947, and are now the most common form, since the discharge mechanism takes advantage of the metastablestate of the inert gas atom to ionize the halogen molecule and produces a more efficient discharge which permits it to operate at much lower voltages, typically 400–600 volts instead of 900–1200 volts. It also has a longer life because the halogen ions can recombine whilst the organic vapor cannot and is gradually destroyed by the discharge process (giving the latter a life of around 108 events).
*, H. J. Spanner, "Gas Filled Tube"
*, G. J. Weissenberg, "Electron Discharge Tube"
*, J. A. Victoreen, "Geiger tube"
*, J. A. Victoreen, "Geiger tube"
* [http://www.alrad.co.uk/electronics/FAQ-GeigerCounters.htm Commercial FAQ sheet on choosing Geiger-Müller tubes]
* [http://www.bookrags.com/sciences/sciencehistory/geiger-counter-woi.html Geiger counter history]
* [http://www.rstp.uwaterloo.ca/manual/detection/gas_filled/gas_filled_detectors.htm Gas filled detectors]
* [http://www-pub.iaea.org/MTCD/publications/PDF/PRTM-1r1_web.pdf IAEA Practical Radiation Technical Manual]
Wikimedia Foundation. 2010.