# Electrostatic force microscope

Electrostatic force microscope

Electrostatic force microscopy (EFM) is a type of dynamic non-contact atomic force microscopy where the electrostatic force is probed. ("Dynamic" here means that the cantilever is oscillating and does not make contact with the sample). This force arises due to the attraction or repulsion of separated charges. It is a long-ranged force and can be detected 100nm from the sample. For example, consider a conductive cantilever tip and sample which are separated a distance $z$ usually by a vacuum. A bias voltage between tip and sample is applied by an external battery forming a capacitor, C, between the two. The capacitance of the system depends on the geometry of the tip and sample. The total energy stored in that capacitor is $U = - frac\left\{1\right\}\left\{2\right\} C Delta V^2$. The battery works to maintain a constant voltage, $Delta V$, between the capacitor plates (tip and sample). By definition, taking the negative gradient of the energy gives the force. The z component of the force (the force along the axis connecting the tip and sample) is thus:

$F_\left\{electrostatic\right\} = frac\left\{1\right\}\left\{2\right\} frac\left\{partial C\right\}\left\{partial z\right\} Delta V^2$.

The electrostatic force can be probed by changing the voltage, and that force is parabolic with respect to the voltage. One note to make is that $Delta V$ is not simply the voltage difference between the tip and sample. Since the tip and sample are often not the same material, and furthermore can be subject to trapped charges, debris, etc., there is a difference between the work functions of the two. This difference, when expressed in terms of a voltage, is called the contact potential difference, $V_\left\{CPD\right\}$ This causes the apex of the parabola to rest at $Delta V = V_\left\{tip\right\} - V_\left\{sample\right\} - V_\left\{CPD\right\} = 0$. Typically, the value of $V_\left\{CPD\right\}$ is on the order of a few hundred millivolts. Forces as small as piconewtons can routinely be detected with this method.

With an electrostatic force microscope, like the atomic force microscope it is based on, the sample can be immersed in liquid.

References

L. Kantorovich, A. Livshits, and M. Stoneham, J. Phys.:Condens. Matter 12, 795 (2000)

* [http://www.pnas.org/cgi/content/abstract/92/22/10384 "Electrostatic Force Microscope for Probing Surface Charges in Aqueous Solutions"] by S. Xu and M.F. Arnsdorf

Wikimedia Foundation. 2010.

### Look at other dictionaries:

• Atomic force microscope — The atomic force microscope (AFM) or scanning force microscope (SFM) is a very high resolution type of scanning probe microscope, with demonstrated resolution of fractions of a nanometer, more than 1000 times better than the optical diffraction… …   Wikipedia

• Magnetic force microscope — MFM images of 3.2 GB and 30 GB computer hard drive surfaces. Magnetic force microscope (MFM) is a variety of atomic force microscope, where a sharp magnetized tip scans a magnetic sample; the tip sample magnetic interactions are detected and used …   Wikipedia

• Kelvin probe force microscope — Kelvin probe force microscopy ( KPFM ), also known as surface potential microscopy, is a noncontact variant of atomic force microscopy (AFM) that was [http://dns.ntu ccms.ntu.edu.tw/references/APPL PHYS LETT 58 2921 1991.pdf invented] in 1991.… …   Wikipedia

• Microscope — This article is about microscopes in general. For light microscopes, see optical microscope. Microscope Us …   Wikipedia

• microscope — /muy kreuh skohp /, n. 1. an optical instrument having a magnifying lens or a combination of lenses for inspecting objects too small to be seen or too small to be seen distinctly and in detail by the unaided eye. 2. (cap.) Astron. the… …   Universalium

• Electrostatic lens — An electrostatic lens is a device that assists in the transport of electrons emitted from a sample to an electron analyzer, analogous to the way an optical lens assists in the transport of light in an optical instrument. The recent development of …   Wikipedia

• Photoconductive atomic force microscopy — (pc AFM) is a scientific technique.. Multi layer photovoltaic cells have gained popularity since mid 1980s.[1] At the time, research was primarily focused on single layer photovoltaic (PV) devices between two electrodes, in which PV properties… …   Wikipedia

• Scanning tunneling microscope — Image of reconstruction on a clean Gold(100) surface …   Wikipedia

• Near-field scanning optical microscope — Near field scanning optical microscopy (NSOM/SNOM) is a microscopic technique for nanostructure investigation that breaks the far field resolution limit by exploiting the properties of evanescent waves. This is done by placing the detector very… …   Wikipedia

• Magnetic resonance force microscopy — (MRFM) is an imaging technique that acquires magnetic resonance images (MRI) at nanometer scales, and possibly at atomic scales in the future. MRFM is potentially able to observe protein structures which cannot be seen using X ray crystallography …   Wikipedia