# Air permeability specific surface

Air permeability specific surface

The air permeability specific surface of a powder material is a single-parameter measurement of the fineness of the powder. The specific surface is derived from the resistance to flow of air (or some other gas) through a porous bed of the powder. The SI units are m2·kg-1 ("mass specific surface") or m2·m-3 ("volume specific surface").

ignificance

When a powder reacts chemically with a liquid or gas at the surface of its particles, the specific surface is directly related to its rate of reaction. The measurement is therefore important in the manufacture of many processed materials. It is universally used in the cement industry as a gauge of product fineness which is directly related to such properties as speed of setting and rate of strength development.

Methods

Measurement consists of packing the powder into a cylindrical "bed" having a known porosity (i.e. volume of air-space between particles divided by total bed volume). A pressure drop is set up along the length of the bed cylinder. The resulting flow-rate of air through the bed yields the specific surface by the Carman equation [Carman P C, "J.Soc.Chem.Ind.", 57, p 225 (1938)] :

:$S=cfrac\left\{7d\right\}\left\{ ho,\left(1-epsilon,\right)\right\}sqrt\left\{dfrac\left\{epsilon,^3pi,delta,P\right\}\left\{leta,Q$

where::S is specific surface, m2·kg-1:d is the cylinder diameter, m:ρ is the sample particle density, kg·m-3:ε is the volume porosity of the bed (dimensionless):δP is the pressure drop across the bed, Pa:l is the cylinder length, m:η is the air dynamic viscosity, Pa·s:Q is the flowrate, m3·s-1

It can be seen that the specific surface is proportional to the square root of the ratio of pressure to flow. Various standard methods have been proposed:
* Maintain a constant flowrate, and measure the pressure drop
* Maintain a constant pressure drop, and measure the flowrate
* Allow both to vary, deriving the ratio from the characteristics of the apparatus.

Lea and Nurse method

The second of these was developed by Lea and Nurse [Lea F M, Nurse R W, "J.Soc.Chem.Ind.", 58, p 227 (1939)] . The bed is 25 mm in diameter and 10 mm thick. The desired porosity (which may vary in the range 0.4 to 0.6) is obtained by using a calculated weight of sample, pressed to precisely these dimensions. The required weight is given by:

:$M= frac\left\{pi\right\}\left\{4\right\},d^2l ho,\left(1-epsilon,\right)$

A flowmeter consisting of a long capillary is connected in series with the powder bed. The pressure drop across the flowmeter (measured by a manometer) is proportional to the flowrate, and the proportionality constant can be measured by direct calibration. The pressure drop across the bed is measured by a similar manometer. Thus the required pressure/flow ratio can be obtained from the ratio of the two manometer readings, and when fed into the Carman equation, yields an "absolute" value of the air permeability surface area. The apparatus is maintained at a constant temperature, and dry air is used so that the air viscosity can be obtained from tables.

Rigden method

This was developed [Rigden P J, "J.Soc.Chem.Ind.", 62, p 1 (1943)] in the desire for a simpler method. The bed is connected to a wide-diameter u-tube containing a liquid such as kerosene. On pressurizing the space between the u-tube and the bed, the liquid is forced down. The level of liquid then acts as a measure of both pressure and volume flow. The liquid level rises as air leaks out through the bed. The time taken for the liquid level to pass between two pre-set marks on the tube is measured by stop-watch. The mean pressure and mean flowrate can be derived from the dimensions of the tube and the density of the liquid.

A later development used mercury in the u-tube: because of mercury's greater density, the apparatus could be more compact, and electrical contacts in the tube touching the conductive mercury could automatically start and stop a timer.

Blaine method

This was developed [Blaine R L, "Bull.Am.Soc.Test.Mater.", 123, p 51 (1943)] independently by R L Blaine of the American National Bureau of Standards, and uses a small glass kerosine manometer to apply suction to the powder bed. It differs from the above methods in that, because of uncertainty of the dimensions of the manometer tube, absolute results can't be calculated from the Carman equation. Instead, the apparatus must be calibrated, using a known standard material. The original standards, supplied by NBS, were certified using the Lea and Nurse method. Despite this shortcoming, the Blaine method is now by far the most commonly used, mainly because of the ease of maintenance of the apparatus and simplicity of the procedure [e.g. ASTM Standard Test Method C 204] .

References

Wikimedia Foundation. 2010.

Нужен реферат?

### Look at other dictionaries:

• Dynamic insulation — is a form of insulation where cool outside air flowing through the thermal insulation in the envelope of a building will pick up heat from the insulation fibres. Buildings can be designed to exploit this to reduce the transmission heat loss (U… …   Wikipedia

• hydrosphere — /huy dreuh sfear /, n. the water on or surrounding the surface of the globe, including the water of the oceans and the water in the atmosphere. [1885 90; HYDRO 1 + SPHERE] * * * Discontinuous layer of water at or near the Earth s surface. It… …   Universalium

• clothing and footwear industry — Introduction also called  apparel and allied industries,  garment industries,  or  soft goods industries,         factories and mills producing outerwear, underwear, headwear, footwear, belts, purses, luggage, gloves, scarfs, ties, and household… …   Universalium

• Soil — For other uses, see Soil (disambiguation). A represents soil; B represents laterite, a regolith; C represents saprolite, a less weathered regolith; the bottommost layer represents bedrock …   Wikipedia

• Earthquake — For other uses, see Earthquake (disambiguation). Global earthquake epicenters, 1963–1998 …   Wikipedia

• Water content — Soil composition by phase: s soil (dry), v void (pores filled with water or air), w water, a air. V is volume, M is mass. Water content or moisture content is the quantity of water contained in a material, such as soil (called soil moisture),… …   Wikipedia

• Building regulations in England and Wales — Building regulations are statutory instruments that seek to ensure that the policies set out in the relevant liegislation are carried out and Building regulations approval is required for most building work in the UK. Building regulations that… …   Wikipedia

• Fouling — This article is about fouling in engineering. For uses of the term foul outside technology, see Foul (disambiguation). Not to be confused with fowling. Heat exchanger in a steam power plant, fouled by macro fouling …   Wikipedia

• Water well — Village pump redirects here. For information on Wikipedia project related discussions see Wikipedia:Village pump. Water well at the Wald monastery in Germany. A water well is an excavation or structure created in the ground by digging, driving,… …   Wikipedia

• Heavy equipment (construction) — Heavy machinery redirects here. For the album by Anders Johansson, Jens Johansson and Allan Holdsworth, see Heavy Machinery (album). Heavy equipment vehicles of various types parking near a highway construction site …   Wikipedia